Secondary power semi-regulated. Secondary electrical power
`forms shall be provided by power supplies which are part of the C&D TS.
Power forms, from 10 percent to full load, and for normal values of line
voltage variations, shall be as listed in Table I.
Secondary power, regulated.
Secondary regulated electrical
power forms shall be provided by power supplies which are part of the C&D TS.
Regulation from no load to full load, and for normal values of line voltage
variations, shall not exceed 2 percent. Ripple voltage for dc power forms
shall not exceed 150 millivolts (mV) peak-to-peak (p-p). The power forms
listed in Table II shall be provided.
The C&D TS shall provide the following interfaces.
Clean air interface. The C&D TS shall interface with the
clean air through ducts which enter the top rear of each test console of the
Electrical power interface, The C&D TS shall interface with
electrical power for the convenience outlets at one end of the convenience
outlet strip mounted on the top of the front face of the C&D TS. The C&D TS
shall interface with all other electrical power through power lines entering
the top of bay number 1.
SECONDARY POWER FORMS, REGULATED
Applicable except as follows.
Electrical power. Applicable.
Electrical power required for
operation of the C&D TS shall be as listed in Table III.