Environmental Test Methods for Aerospace and
AF Drawing 69F32621
AF Drawing 69B32697
(Copies of specifications, standards, drawings, and publications required by
suppliers in connection with specific procurement functions should be obtained
from the Procuring Activity or as directed by the Contracting Officer).
First Article tests of three Test Sets shall be made in
accordance with the provisions contained herein.
The requirements of MIL-T-21200 for Class 2
3.2 General Specification.
equipment are applicable as requirements of this specification with the
exceptions or additions called out herein.
Test Set identification conforming to
3.2.1 Nameplate Identification.
Approval of the drawing for the
MIL-STD-130 shall be provided on the chassis.
nameplate shall be by the Engineering Division of the Procuring Activity.
The equipment shall operate with no degradation of
3.3 Service Conditions.
performance under any combination of service conditions specified in
The design criteria shall be those applicable
MIL-STD-810, Table 1, Column E.
to Ground Equipment and worldwide operation.
The dimensions of the Test Set and cables shall be in
accordance with AF Drawing No. 69F32621.
AF Drawing No. 69F32621 shall be used as a reference
3.4.2 Design Reference.
and guide for design of the Test Set.
The Test Set shall comprise one complete unit which shall
include the chassis with control panel, cable assembly and attaching
Selection of components and design criteria
provisions for an AN/PSM-6 Meter.
shall be in accordance with the provisions of MIL-T-21200.
3.4.4 Attaching Provisions.
Two catches shall be provided on the chassis
Two leads with connectors
bracket for physically attaching an AN/PSM-6 Meter.
shall be provided for electrical connection to the AN/PSM-6 Meter.
The Test Set shall be designated for use with
3.4.5 Electrical Requirements.
voltages up to 115 Volts Alternating Current (VAC).