4.4.2 Failure of sample.- Should any one item of a special sampling
fail to meet the specified test requirements, acceptance of the represented
inspection lot will be suspended by the Government until necessary correc-
tions have been made by the contractor and the resubmitted samples have
been approved (see 4.3.3).
4.5 Inspection equipment.- Except as otherwise provided for by the
contract, the contractor shall supply and maintain inspection equipment
in accordance with the applicable requirements of MIL-I-45607.
4.5.1 Government furnished inspection equipment.- Where the contract
provides for Government furnished test equipment, supply and maintenance
of test equipment shall be in accordance with the applicable requirements
specified in MIL-I-45607.
4.5.2 Contractor furnished inspection equipment.
188.8.131.52 Government design.- unless otherwise specified in the contract,
all inspection equipment specified by drawing number in specifications or
SQAP forming a part of the contract shall be supplied by the contractor in
accordance with technical data listed in the List of Inspection Documents
when provided with the Technical Data Package (TDP).
184.108.40.206 Contractor design.- The contractor shall design and supply
inspection equipment compatible with the "Test Methods and Procedures"
specified in 4.6 of this specification and with the component inspection
procedures specified in "Examination" and "Test Facilities" requirements
of MIL-F-13926. Since tolerance of test equipment is normally considered
to be within 10% of the product tolerance for which it is intended, this
inherent error in the test equipment design must be considered as part
of the prescribed product tolerance limit. Thus, concept, construction,
materials, dimensions and tolerance used in the design of test equipment
shall be so selected and controlled as to insure that the test equipment
will reliably indicate acceptability of a product which does not exceed
90% of the prescribed tolerance limit, and permit positive rejection when
non-conforming. Construction shall be such as to facilitate routine
calibration of test equipment.
4.5.3 Test equipmnent.- In conjunction with 220.127.116.11, the following
standard test equipment shall be utilized in the performance of the
applicable tests as specified in 4.6.
Standard type conforming to the accuracies
1. Hot and cold chamber
outlined in test facilities specified
under test facilities of MIL-F-13926.
Standard shop type accurate to within
2. Torque wrench and scale
10% of drawing tolerance.
4.6 Test methods and procedures.