Requirements and test procedures.
Torque (+150 F & -5O F)
4 . 4 . 2 Failure o f
Should any one item of a special
sampling fail to meet the specified test requirements, acceptance of
the represented inspection lot will be suspended by the government
until necessary correction have been made by the contractor and the
resubmitted sampling have been approved (See 4.3.3) .
4.5 Test equipment.
Except as otherwise provided for
by the contract, the contractor shall supply and maintain inspection
equipment in accordance with the applicable requirements of
furnished inspection equipment. Where the
contract provides for Government furnished test equipment, supply
and maintenance of test equipment shall be in accordance with the
applicable requirements specified in MIL-I-45607.
126.96.36.199 Government desiqn. Unless otherwise stated in the
contract, all inspection equipment specified by drawing number in
specifications or QAP forming a part of the contract shall be
supplied by the contractor in accordance with technical data listed
in the List of Inspection Documents when provided with the Technical
Data Package (TDP).
188.8.131.52 Contractor design. The contractor shall design and
supply inspection equipment compatible with the requirements of
Since tolerance of test equipment is normally
considered to be within 10% of the product tolerance for which it is
intended, this inherent error in the test equipment design must be
considered as part of the prescribed product tolerance limit. Thus ,
concept, construction, materials, dimensions, and tolerance used in
the design of test equipment shall be so selected and controlled as
to insure that the test equipment will reliably indicate accept-
ability of a product which does not exceed 90% of the prescribed
tolerance limit, and permit positive rejection when non-conforming.
Construction shall be such as to facilitate routine calibration of
Privacy Statement - Press Release - Copyright Information. - Contact Us