zero to 1600 roils (with respect to a horizontal plane) within 0.40
3.7 Torque.- The running torque required to rotate the micrometer- .
knob shall not exceed 2.0 pound-inches at the operating temperatures
specified in 3.4.2.
3.8 Level vial.- The level vial bubble shall be visible in all
ambient light conditions
QUALITY ASSURANCE PROVISIONS
4.1 Responsibility for inspection.- Unless otherwise specified in
the contract or purchase order, the supplier is responsible for the
performance of all inspection requirements as specified herein. Except
as otherwise specified, the supplier may utilize his own facilities or
any other facilities suitable for the performance of the inspection re-
quirements specified herein, unless disapproved by the Government. The
Government reserves the right to perform any of the inspection set
forth in the specification where such inspections are deemed necessary
to assure supplies and services conform to prescribed requirements.
4.1.1 General provisions.- The component and subassembly inspec-
tion requirements of MIL-F-13926 form a part of the quality assurance
provisions of this specification. Definitions of inspection terms shall
be as listed in MIL-STD-109.
4.2 First article (initial production) approval.- The requirement
for first article approval and the responsibility (Government or contractor)
for first article testing shall be as specified in the contract. The
sample for first article approval tests shall consist of three gunner's
quadrants plus three (3) each of all items covered by SQAP. The sample
shall be manufactured in the same manner, using the same materials
equipment, processes, and procedures as used in regular production. All
parts and materials, including packaging and packing, shall be obtained
from the same source of supply as used in regular production.
When the Government is responsible for
4.2.1 Government testing.-
conducting first article approval tests, the contractor, prior to sub-
mitting the sample to the Government, shall inspect the sample to in-
sure that it conforms to all the requirements of the contract and sub-
mit a record of this inspection with the sample, including certificates
of conformance for materials.
4.2.2 Contractor testing.- When the contractor is responsible for
conducting first article approval tests, the sample shall be inspected
by the contractor for all the requirements of the contract. The sample
Privacy Statement - Press Release - Copyright Information. - Contact Us